Home»Projects »Xilinx Logic Analyzer Adapter
November 10, 2010
January 14, 2009
October 9, 2007
November 12, 2005
February 21, 2005
September 1, 2005
April 28, 2006
September 29, 2001
The Intronix Logicport is a small USB-based logic analyzer. It has 34 channels, two of which can be used as clock inputs, and does state analysis at 200 MHz. Sample depth without compression is 2048 samples per channel. While the Logicport is modest in specifications, for many tasks it may well be all you need. I mainly use the Logicport for Xilinx FPGA projects.
The Logicport connects to the system under test using one of three ways:
Logicport logic analyzer with flying leads. To the right, the Mictor adapter.
Plugging 38 color-coded leads on 38 header pins on my Xilinx Development Board is time-consuming and error-prone.
Instead of connecting through a flying lead probe, connect through a custom adapter.Benefits:
|Adapter, front view.||Adapter, back view.|
Adapter sitting on a Xilinx ML403, using the Xilinx Generic Interface (XGI) connector. Other Xilinx developments boards feature the same connector. Please note the XGI connector is non-polarized. Make sure you do not plug in the adapter upside-down. Plugging in the adapter upside-down will short some FPGA outputs to ground, possibly damaging them.
This setup allows you to look at 32 signals, a clock enable and a clock inside the FPGA in real time at reasonable cost.
When connecting the Logicport to a circuit, you are connecting the circuit ground to the PC ground through the USB cable. If circuit ground and PC ground have a different potential, connecting them produces a ground loop. This is a general problem, which may also occur in USB oscilloscopes, for instance.
A simple solution against ground loops, sufficient for occasional use, is using a notebook with the AC cord unplugged, running off the battery. My personal choice has been to put a 100VA isolation transformer Block TIM-100 between notebook and mains and forgetting all about ground loops. Connecting the notebook to the network using unshielded twisted pair ethernet does not break the isolation.
I neither manufacture nor sell these adapters. If you want one, you will have to make one yourself. If you wish to have the printed circuit board made for you, download the archive for the adapter and extract all files. Send the .brd file to a PCB house such as pcb-pool. The .brd file is an Eagle format file. Ask for a double-sided PCB with solder mask, no silkscreen, no electrical testing.
The schematics and board layout can be downloaded:
You can use the sample ISE project as a simple electrical test. You should get the following waveform:
Last update page: October 9, 2007